1. Designed for Large-level Lab.
2. Dual methodologies: Cone plate method, Capillary method.
3. Dual Sample Plates: Whole blood and plasma can be performed simultaneously.
4. Bionic Manipulator: Reversal mixing module, mixing more thoroughly.
3. External barcode reading, LIS support.
4. Non-Newtonian standard marker win China National Certification.
|Test principle||whole blood test method: cone-plate method; plasma test method: cone-plate method, capillary method;|
|Working mode||dual needle dual disk, dual methodology dual test system can work in parallel at the same time|
|Signal acquisition method||Cone plate signal acquisition method adopts high-precision grating subdivision technology; Capillary signal acquisition method adopts self-tracking liquid level differential acquisition technology;|
|Movement material||titanium alloy|
|Test time||whole blood test time ≤30 seconds/sample, plasma test time ≤1 second/sample;|
|Viscosity measurement range||(0～55) mPa.s|
|Shear stress range||(0～10000) mPa|
|Range of shear rate||(1～200) s-1|
|Sample amount||whole blood ≤800ul, plasma ≤200ul|
|Sample position||double 80 holes or more, fully open, interchangeable, suitable for any test tube|
|Instrument control||use workstation control method to realize instrument control function, RS-232, 485, USB interface optional|
|Quality control||It has the non-Newtonian fluid quality control materials registered by the National Food and Drug Administration, which can be applied to the non-Newtonian fluid quality control of the bid products, and can be traced to the national non-Newtonian fluid standards.|
|Scaling function||the non-Newtonian fluid viscosity standard material produced by the bidding product manufacturer has obtained the national standard material certificate|
|Report form||open, customizable report form, and can be modified on site|
Cone-plate: complete measurement range, pointwise, prompt, steady state method.
Capillary: micro capillary prompt method (pressure sensor).
3. Signal collection technology: High-precision raster subdivision technology.
4. Working mode: simultaneous working with dual-cap piercing probe (with liquid level sensor function), dual-sample plate, dual-methodologies, three testing modules can work simultaneously.
5. Cap-Piercing function: sample cap-piercing probe module for capped sample tube.
B. Working environment:
1. Operating voltage: 100~240 VAC, 50~60 Hz.
2. Power consumption: 350 VA.
3. Operating temperature: 10~30 °C.
4. Humidity: 30~75%.
C. Working parameters:
1. Accuracy: Newtonian fluid <±1%. Non Newtonian fluid <±2%.
2. CV: Newtonian fluid ≤1%. Non Newtonian fluid ≤2%.
3. Throughput: ≤30 s/sample (whole blood). ≤0.5 s/sample (plasma).
4. Shear rate range: (1~200) S-1.
5. Viscosity range: (0~60) mPa·s.
6. Shear force range: (0~12000) mPa.
7. Sampling volume: 200~800